Automation compatible – Secure handling – Easy observation
The one-and-only 96-well spheroid plate with InSphero’s unique features specially designed for lossless spheroid/organoid formation and handling as well as reliable culturing and analysis.
- Unique SureXchange™ ledge engineered to prevent unintended spheroid/organoid aspiration
- Spheroid/organoid morphology and functionality is preserved in long-term culture by unique ULA coating
- Enhanced imaging speed and resolution through flat bottom well
- Improved spheroid/organoid formation with the Akura™ Tilting Stand by keeping plate at 30 degrees during cell aggregation
20-pack of individually wrapped Akura™ 96 Spheroid Microplates, made of polystyrene, high-efficiency ULA coating, sterile, includes lid
Shipping time: 4 business days
- Conical well design with SureXchange™ ledge protecting the spheroid/organoid from aspiration while leaving a minimal residual volume (5-7 µl).
- ANSI/SLAS standard format for full compatibility with automated liquid handling systems
- Leak-free and glue free one-component transparent polystyrene plate
- >90% medium removal without spheroid loss using manual and automated pipetting
- Ultra-low attachment surface with stable cell-repellent coating
- Biologically inert, and non-degradable surface compatible with cell culture and in-plate post-culture processing steps such as cell lysis and fixation procedures
- Flat bottom well minimizing optical aberration and ensuring constant spheroid z-position
- Defined 1-mm quick-reference region of interest for increased imaging speed
At InSphero we are using our Akura™ Spheroid Microplates daily to produce and assay 3D microtissues for our partners in pharma and biotech. Our manuals and protocols contain valuable information and hands-on advice to make sure that these great plates work in your hands perfectly.
- Product Manual Akura™ 96 Spheroid Microplate
- Technical Specifications Akura™ 96 Spheroid Microplate
- Quick Start Guide Akura™ 96 Spheroid Microplate
- Certificate of Compliance
A detailed protocol for production of spheroids in the Akura™ 96 Plate is provided in the product manual. Below are answers to some frequently asked questions to help get you started.
Q: Why do you recommend pre-wetting of the wells prior to spheroid/organoid seeding?
A: Pre-wetting the wells of the Akura™ 96 Plate is required prior seeding to prevent inclusion of air-bubbles. For that, apply 40 µl of your cell medium to each well by placing the tips far into the wells. Remove the pre-wetting solution by placing the tip at the ledge of the upper cavity of the well. Aspirate medium until it is completely removed from each well. A negligible amount (< 5-7 µl) may remain in the bottom of the chamber.
Q: Could you recommend a cell concentration for my cell suspension for the generating of spheroids/organoids?
A: For long-term growth profiling, we recommend starting with low cell numbers (250 – 500 cells per well of 70 µl). If use of non-proliferating cells or rapid production of larger spheroids are required, start with higher numbers (from 2500+ cells per 70 µl). Generally, we recommend trying different concentrations for defining your optimal range when using new cell types.
Q: What is the optimal volume per well in the Akura™ 96 Plate?
A: To achieve an optimal and stable hanging drop, gently deliver 70 µl (pipetting speed < 10 µl/sec) of cell suspension into each well of the Akura™ 96 Plate by placing the pipette tips near, but not touching, the bottom of the wells.
Important - For spheroids with uniform size and cell composition it is essential to assure a homogeneous distribution of the cells by gently pipetting up and down prior to seeding into the Akura™ 96 Plate.
Q: Why do you recommend centrifuging the Akura™ 96 Plate after cell seeding?
A: We recommend to briefly centrifuge the plate after cell seeding to remove any air bubbles and to force the cells to the bottom of the well in order to promote cell-aggregation and spheroid formation.
For that, place the lid on the plate and spin in a microtiter-plate centrifuge for 2 minutes at 250 RCF. Afterwards, incubate the plate in a humidified CO2 incubator at 37 °C for 2-5 days.
Q: How do I exchange the medium in the Akura™ 96 Plate without disturbing or losing the spheroids?
A: To prevent spheroid/organoid loss during the exchange of media, the SureXchange™ ledge at the inside wall of each well serves as an anchoring point for the pipette tip. Just place the tip at the ledge of the well, see figure below, and remove the medium at low pipetting speed (>30 µl/sec). A minimal volume of ~ 5-7 µl will remain in the well.
Then, add 70 µl of fresh medium by placing the pipette tip at the ledge, use dispensing rate <50 µl/sec.
Important - when using automated liquid handling devices, an off-center alignment of the vertical pipette tip will achieve the same effect.
Q: What is the best way to prevent evaporation in the outer wells of my plates?
A: Evaporation in the outer (perimeter) rows of wells is a phenomenon common to most low-volume culture platforms, and thus requires careful attention to maintaining proper humidity control. If not controlled, pronounced evaporation can result in concentration or precipitation of media components (serum, salt) that can impact spheroid formation or health, and can alter the effective concentration of a compound/additive in the medium over the course of a long-term experiment. To provide maximum humidity control when using the Akura™ Plates, we recommend the following:
- Use an incubator with good humidity control (>95% of rel. humidity), and exercise best practice in maintaining and minimizing loss of humidity (e.g. minimize incubator door opening and closing).
- For culture in the Akura™ 96 Plate, at least 50-70 µl of medium in each well is recommended and can be increased to a maximum of 80 µl if incubator humidity control is a persistent issue. Medium exchange frequency can also be increased to every other day or daily if conditions dictate.
- We recommend the use of the InSphero Incubox to reduce edge effects when performing long-term culture with low-frequency medium exchange. The Incubox will be launched in Q4/2021 and will be available in shop.insphero.com